Contact heat testing device CHD DIN EN ISO 12127-1
- Determining the throughput of contact heat through protective clothing or its materials.
The device determines the time threshold of a temperature rise of 10°C in the calorimeter. In the test a heating cylinder is heated to a defined temperature and the specimen is put on the calorimeter. During the measuring the calorimeter is moved towards the heating cylinder with a defined skipping speed of 5 mm/s and after contact a load of 45N is applied. The temperature of the heating cylinder is adjustable from room temperature up to 500 °C. The power supply is provided by an IEC-320 AC power cordset 230 VAC (50Hz).
|Geometry of specimen:||Textiles|
|Testing standards:||DIN EN ISO 12127-1|
|Sensory functions:||Temperature (calorimeter)|
|Power input:||230 VAC / 350 VA|
|Dimensions (testing device):||600 x 250 x 650 mm (W x D x H)|
|Weight (testing device):||ca. 30 kg|