Testing device HBP DIN EN ISO 6942
- Assessment of materials and material combinations that are exposed to a heat radiation source.
Procedure A exposes a specimen in a free-standing frame to a defined heat radiation for a defined time period. The intensity of the heat radiation is controlled by the distance between specimen and heat radiation source. After the testing the specimen and its different layers are examined for visual changes.
In procedure B the specimen is fixed onto a calorimeter and exposed to a defined heat radiation. The time needed for a temperature rise in the calorimeter of 12°C and 24°C respectively is measured.
On the basis of this data the heat throughput rate is calculated from the difference between passed and incident heat flow density. The software includes the calculations and evaluation of the standard. The power supply is provided by a three-wire plug 3 ~ 400 VAC (50/60 Hz) (5x32 A 6h CEE). The cooling mechanism works with a continuous water flow which is provided by an inlet and outlet on the testing device.
Data recording via control computer
Software guided calibration and test procedures.
Calorimeter protection by pressure switch preventing operation without co
|Type of specimen:||Textiles|
|Test standards:||DIN EN ISO 6942|
|Sensors:||Temperature (calorimeter), water pressure sensor|
|Supplies:||3 ~ 400 VAC / 12.5 kVA, Cooling water, waiste water (sink sufficient)|
|Dimensions:||2000 x 800 x 1500 mm (W x D x H)|
|Weight:||approx. 100 kg|
|Requirements for the installation site:||Waiste water connection|